Specification of the four DOSTEL detectors. The thickness (column 2), area (column 3), mass (column 5), and energy deposition ranges (column 5) are shown. For the telescope detectors these energy ranges can be converted to LET in water.
|LET range/(keV μm−1 in water)|
|Upper tele. detector||154||6.93||0.249||dEi = [28.7,…, 22612]|
|LETi = [0.084,…, 66.3]|
|Lower tele. detector||315||6.93||0.509||dEi = [42.5,…, 51248]|
|LETi = [0.061,…, 73.5]|
|Pin diodes||300||2.31||0.161||dEi = [44.4,…, 56292]|
|no pathlength limitation|
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.